No reviews found.
No annotations yet.
|Metadata||Start Date||End Date||Comment|
Glider optodes typically exhibit a sensor lag of 20-25 seconds. Example attached of several segments with consecutive down and up casts for CP05MOAS-GL335-04-DOSTAM000 on 10/26/2015 showing a typical measurement time lag. Profiles should be analyzed for individual time lags and data should be adjusted. OOI doesn't necessarily need to make the adjustment but users should be aware of the issue.
By Lori Garzio, on 5/31/19
|7/5/16, 12:00 AM||7/5/16, 12:00 AM||
Deployment 1 - glider deployed on 06/27/2016 and recovered on 07/5/2016 due to a leak in the forward compartment.
By Lori Garzio,