|Dep.||Preferred Method||Stream||DD||FD||SG||EG||Gaps||GD||TS||Rate (s)||Pressure Comp.||Time Order||Valid Data||Missing Data||Data Comp.||Missing Coords.||Review|
|1||telemetered||2||2||0||0||0||0||382||60||1000 / 1005||1||1||Complete|
No annotations yet.
|Metadata||Start Date||End Date||Comment|
Glider optodes typically exhibit a sensor lag of 20-25 seconds. Example attached of several segments with consecutive down and up casts for CP05MOAS-GL335-04-DOSTAM000 on 10/26/2015 showing a typical measurement time lag. Profiles should be analyzed for individual time lags and data should be adjusted. OOI doesn't necessarily need to make the adjustment but users should be aware of the issue.
By Lori Garzio, on 5/31/19
Short (2 day) deployment and only telemetered data available for download - dataset should be annotated.
By Lori Garzio, on 5/30/19