|recovered_host||ctdgv_m_glider_instrument_recovered||Data Products||Report M2M Stats||Science|
|telemetered||ctdgv_m_glider_instrument||Data Products||Report M2M Stats||Science|
|Deployment||Cruise||Start Date||Stop Date||Mooring Asset||Node Asset||Sensor Asset||Latitude||Longitude||Deployment Depth||Water Depth|
|Metadata||Start Date||End Date||Comment|
Glider optodes typically exhibit a sensor lag of 20-25 seconds. Example attached of several segments with consecutive down and up casts for CP05MOAS-GL335-04-DOSTAM000 on 10/26/2015 showing a typical measurement time lag. Profiles should be analyzed for individual time lags and data should be adjusted. OOI doesn't necessarily need to make the adjustment but users should be aware of the issue.
By Lori Garzio, on 5/31/19
Deployment 1 should be annotated to indicate why the deployment was so short and why recovered data are not available.
By Lori Garzio, on 5/30/19
No annotations yet.