|recovered_host||cg_dcl_eng_dcl_cpu_uptime_recovered||CPU Uptime Data||Report M2M||Engineering|
|recovered_host||cg_dcl_eng_dcl_dlog_aarm_recovered||Accelerometer Data||Report M2M||Engineering|
|recovered_host||cg_dcl_eng_dcl_dlog_mgr_recovered||Data Log Manager||Report M2M||Engineering|
|recovered_host||cg_dcl_eng_dcl_dlog_status_recovered||Data Logger Status||Report M2M||Engineering|
|recovered_host||cg_dcl_eng_dcl_error_recovered||Error Data||Report M2M||Engineering|
|recovered_host||cg_dcl_eng_dcl_gps_recovered||GPS Data||Report M2M||Engineering|
|recovered_host||cg_dcl_eng_dcl_msg_counts_recovered||Message Counts Data||Report M2M||Engineering|
|recovered_host||cg_dcl_eng_dcl_pps_recovered||Pulse Data||Report M2M||Engineering|
|recovered_host||cg_dcl_eng_dcl_status_recovered||Status Data||Report M2M||Engineering|
|recovered_host||cg_dcl_eng_dcl_superv_recovered||Leak Detect Data||Report M2M||Engineering|
|telemetered||cg_dcl_eng_dcl_cpu_uptime||CPU Uptime Data||Report M2M||Engineering|
|telemetered||cg_dcl_eng_dcl_dlog_aarm||Accelerometer Data||Report M2M||Engineering|
|telemetered||cg_dcl_eng_dcl_dlog_mgr||Data Log Manager||Report M2M||Engineering|
|telemetered||cg_dcl_eng_dcl_dlog_status||Data Logger Status||Report M2M||Engineering|
|telemetered||cg_dcl_eng_dcl_error||Error Data||Report M2M||Engineering|
|telemetered||cg_dcl_eng_dcl_gps||GPS Data||Report M2M||Engineering|
|telemetered||cg_dcl_eng_dcl_msg_counts||Message Counts Data||Report M2M||Engineering|
|telemetered||cg_dcl_eng_dcl_pps||Pulse Data||Report M2M||Engineering|
|telemetered||cg_dcl_eng_dcl_status||Status Data||Report M2M||Engineering|
|telemetered||cg_dcl_eng_dcl_superv||Leak Detect Data||Report M2M||Engineering|
|Deployment||Cruise||Start Date||Stop Date||Mooring Asset||Node Asset||Sensor Asset||Latitude||Longitude||Deployment Depth||Water Depth|
|Metadata||Start Date||End Date||Comment|
|10/12/17, 12:00 AM||6/14/18, 8:02 AM||
The surface buoy detached from the mooring and the subsurface portion of the mooring fell to the seafloor (see annotation ID 102). Excluding all mooring data for this time range from final statistics.
By Lori Garzio, on 1/15/19
|Metadata||Start Date||End Date||Comment|
||2/14/15, 7:00 PM||8/21/15, 8:00 PM||
Deployment 1: Status data sent from the buoy included leak detects in the buoy well, drop in battery voltage, and loss of wind turbine input. Upon recovery, the buoy was primarily intact but several instruments were damaged and/or missing. Ice build-up on the tower is speculated to be the cause for much of the damage. No telemetered or recovered_host data expected. Functional instruments could continue to collect data using internal battery power and storage cards.
Id: 46 By: lgarzio
||1/26/16, 7:00 PM||7/18/16, 8:00 PM||
Deployment 2: A period of violent weather caused power outages on multiple instruments. No telemetered or recovered_host data expected. Functional instruments could continue to collect data using internal battery power and storage cards. Upon recovery, the buoy well was flooded.
Id: 47 By: lgarzio
||12/26/16, 7:00 PM||4/6/17, 8:00 PM||
Deployment 3: Wind turbines were disabled due to evidence of ice build-up on the surface buoy tower. The platform's battery charge state declined steadily. As of 2017-01-15, all DCLs were shut down and the telemetry schedule was reduced to conserve power. Battery-powered instruments should continue to collect data. UPDATE 2017-04-07: With increasing daylight at this location, sufficient power was generated by solar panels to re-establish limited high-bandwidth telemetry and data transmission. UPDATE: at recovery, wind turbines were missing their hubs and blades, and one solar panel was missing.
Id: 121 By: lgarzio
||8/5/17, 2:17 PM||6/14/18, 4:02 AM||
Deployment 4: at 10:03 UTC on 12 October 2017 the Irminger Sea Surface Mooring stopped all communications. At recovery, the surface buoy was missing. The subsurface portion of the mooring had fallen to the seafloor and was recovered. CTDMOs, CTDBP-F, DOSTA, and ADCP instrumentation were full ocean depth rated and survived, although any recovered data available after 2017-10-12T10:03:00 are suspect. All other instrumentation imploded or flooded. The cause for the loss of the buoy is inconclusive.
Id: 102 By: lgarzio
||6/19/19, 8:00 PM||
Deployment 5: Battery voltage dropped below 24V. All DCLs were turned off. CPM1 was placed on a wake schedule. Data collection and telemetry will be irregular until voltage rebounds.
Id: 1625 By: cdobson
|10/21/19, 8:00 PM||
Deployment 6: *UPDATED 2020-04-27: As of 2020-02-21 data for most instruments are being telemetered again (via Iridium). A gap in telemetered data is expected between 2019-10-22 and 2020-02-21 and will not be backfilled but these data will be available upon recovery.*Both fleet broadband units have failed and no more telemetered data is expected at this time. Instruments are believed to be operating as normal and their data will be available upon recovery.
Id: 1743 By: cdobson